How to Use the Tes 3600 Power Analyzer Software 48 to Measure and Analyze Power Systems
The Tes 3600 Power Analyzer Software 48 is a powerful tool that allows you to measure and analyze various parameters of power systems, such as voltage, current, power, power factor, phase angle, frequency, energy, harmonic distortion and waveform. The software works with the Tes 3600 Power Analyzer, a portable device that can connect to four current sensing clamps and measure single-phase or three-phase power systems. In this article, we will show you how to use the software to perform different tasks and get the most out of your power analyzer.
To install the software, you need a PC with Windows XP or later operating system and a USB port. You also need the software CD-ROM and the optical USB interface that came with your power analyzer. Follow these steps to install the software:
Insert the CD-ROM into your PC's CD drive and run the setup.exe file.
Follow the on-screen instructions to complete the installation.
Connect the optical USB interface to your PC's USB port and to the power analyzer's RS-232 port.
Turn on the power analyzer and press the DATA key until you see \"PC\" on the display.
Launch the Tes 3600 Power Analyzer Software 48 from your PC's start menu or desktop icon.
Using the Software
The software has four main functions: data logging, data reading, waveform display and harmonic analysis. You can access these functions from the main menu or the toolbar of the software. Here is a brief overview of each function:
Data logging allows you to record and save the measurements from the power analyzer to your PC in real time. You can set the logging interval from 1 second to 60 minutes and choose which parameters to log. You can also view the logged data in a table or a graph format and export them to a CSV file for further analysis.
Data reading allows you to read and display the data that you have manually stored in the power analyzer's memory. You can store up to 50 sets of data in the power analyzer by pressing the MEM key when measuring. You can then read these data from the software and view them in a table or a graph format. You can also export them to a CSV file for further analysis.
Waveform display allows you to view and analyze the voltage and current waveforms of each phase of the power system. You can select which phase to display and zoom in or out of the waveform. You can also measure the peak-to-peak value, RMS value, frequency and duty cycle of the waveform.
Harmonic analysis allows you to measure and display the harmonic distortion of each phase of the voltage and current waveforms. You can view up to 63 orders of harmonics and their percentage of total harmonic distortion (THD). You can also view the harmonic spectrum in a bar chart or a pie chart format.
The Tes 3600 Power Analyzer Software 48 is a useful tool that enhances the functionality of your power analyzer. It allows you to measure and analyze various aspects of power systems with ease and accuracy. You can also save, export and share your data for further processing or reporting. If you have any questions or problems with using the software, please refer to the user manual or contact us for support. 0efd9a6b88